Professional Manufacturer of Rapid Test Devices
Payment & Shipping Terms:
|Specimen:||Serum, Plasma, Whole Blood||Methodology:||Colloidal Gold|
|Shelf Life:||2 Years||Format:||Cassette|
|OEM / ODM:||Available||Result Time:||Read In 20 Minutes|
25ml Plasma HIV Rapid Test Kit,
100% HIV Rapid Test Kit,
25ml Plasma HIV test Cassette
HIV(1+2) Antibody Rapid Test Kit
For qualitative detection of HIV(1+2) antibodies in serum/plasma and whole blood
☀ Sensitivity: 100%
☀ Specificity: Higher than 99%
☀ Simple: No Instrument Required
☀ Ambient Storage
☀ Reliable: able to differentiate HIV Type I and Type II
☀ Certified by Authoritative Certification
☀ Unique 3-line Patented Design
☀ Winner of "the 2008 National HIV Antigen Diagnostic Kit for Clinical Performance Assessment"
1. HIV-1(gp41 and gp120) and HlV-2(gp36) specific recombinant antigens are separately precoated on to the membrane inzone1and2as the capture reagent on the test zone.
2. During the test, specimen is allowed to react with the colloidal gold particles, which have been conjugated with HIV-1andHIV-2 specific recombinant antigens.
3. Antibodies to HIV-1 and/or HIV-2, if present, will specifically bind to colloidal gold-antigen complex.
4. When the colloidal gold-antigen-antibody complexes move to the test zone, they will specifically bind to the precoated antigens. At the same time, a red colored line will develop in zone 1 and/or 2 on the membrane. Absence of these red colored lines in the test zone ( 1 and 2) suggests a negative result.
5. To serve as a procedural control, red colored line in the control zone will always appear regardless of the presence of antibodies to HIV-1/HIV-2.
Reagents and Materials Provided
Each kit contains:
☀ 40 test cassettes(individually pouched)
☀ Each pouch contains one cassette with one desiccant bag
☀ One bottle of diluent buffer(5ml)
☀ 40 disposable plastic droppers
☀ Instruction for use
Materials Required But Not Provided
☀ Timer or stopwatch
☀ Blood collection devices, for the testing of venous whole blood, serum or plasma
☀ Biohazard disposal container
☀ Disposable gloves
For finger stick samples, the following materials are required:
☀ Alcohol pad
☀ Sterile lancet
☀ Sterile gauze or cotton
☀ Place the test cassette on flat surface. Before unseal the pouch, allow the test cassette to reach room temperature(4-30°C) .Use it immediately once unsealed.
☀ Open the pouch and add1drop(30-40uL) of specimen into the sample well(s) .
☀ When the specimen is completely absorbed, slowly add1drop(45-55pL) of diluent buffer vertically into the sample well(s) .
☀ Avoid dropping specimen or diluent buffer in the observation window.
☀ Do not allow the diluent buffer bottle touch the sample well when dropping the diluent buffer so as to prevent the cross contamination with the specimen.
☀ Observe the result between 15-30 minutes after the diluent buffer added.
Interpretation of Results
Note:‘C'-Control line； ‘1'-HIV-1 Test line； ‘2'-HIV-2 Test line
Negative: No redlines appear within 30 minutes in the test zone( 1 and 2), only a redline in the control zone(C) , which indicates that no antibodies to HIV 1+ 2 have been detected with this test. However, this does not exclude the possibility from infection with HIV.
Positive: One redline in the control zone(C) and one or two red visible lines of any intensity in the test zone( 1 and 2) .This indicates the specimen contains HIV-1 and/or HIV-2 antibodies.
Invalid: No redlines appear in the control zone(C), regardless of whether there is a redline in the test zone( 1 and 2) , indicating that the test is invalid. Discard the test cassette and perform with new cassette.
Newscen HIV(1+2) Antibody Rapid Test has a built-in procedural control that demonstrates assay validity. A redline appeared on the control zone(C) indicates that the test runs correctly.
For Invitro Diagnostic Use ONLY
Read the package insert completely before use. It is very important that the correct procedure is followed. Fail to add the patient sample may lead to a false negative result (i.e.a missed positive) .
Newscen HIV(1+2) Antibody Rapid Test should be stored at room temperature(4-30℃, do not freeze)
for 24 months from the date of manufacture. Keep the test cassette in sealed pouch until use. Once you have taken the test cassette out of the pouch, perform the test as early as possible(within 1hour) to avoid test cassette from becoming moist. Do not use the test beyond the indicated expiration date.
The diluent buffer should be stored at room temperature(4-30℃, do not freeze)
Contact Person: Mr. Randy.Zhang