Professional Manufacturer of Rapid Test Devices
|Place of Origin:||China|
|Certification:||ISO9001, CE, TUV, SFDA|
|Minimum Order Quantity:||10 Boxes or 400 Kits (40 Kits/Box)|
|Price:||Negotiable as per Order Quantity|
|Packaging Details:||40 Kits/Box, 50 Boxes/Carton, 317g/Box, 18kgs/Carton, Box Size: 250x125x65mm, Carton Size: 650x510x330mm|
|Delivery Time:||8 days|
|Payment Terms:||T/T, Western Union, MoneyGram|
|Supply Ability:||100,000 Kits Per Day|
|Specimen:||Serum, Plasma, Whole Blood||Methodology:||Colloidal Gold|
|Shelf Life:||2 Years||Format:||Cassette|
|OEM / ODM:||Available||Result Time:||Read In 20 Minutes|
In Vitro Diagnostic HIV test Cassettes,
TUV HIV test Cassettes,
FDA HIV test Cassettes
HIV(1+2) Antibody Rapid Test Kit
For qualitative detection of HIV(1+2) Antibodies in serum/plasma and whole Blood
► Sensitivity: 100%
► Specificity: Higher than 99%
► Simple: No Instrument Required
► Unique 3-line Patented Design
► Ambient Storage
► Reliable: able to differentiate HIV Type I and Type II
► Certified by Authoritative Certification
► Winner of "the 2008 National HIV Antigen Diagnostic Kit for Clinical Performance Assessment"
The HIV-1/2 Rapid Diagnostic Test (RDT) is a qualitative test for the detection of antibodies to Human Immunodeficiency Virus type 1 and 2 (HIV-1/2) in human serum/plasma.
It is considered as an initial screening test for HIV-1/2 antibodies. All positive specimens must be confirmed with Western Blot or other qualified EIA.
Reagents and Materials Provided
Each kit contains:
► 40 test cassettes(individually pouched)
► One bottle of diluent buffer(5ml)
► 40 disposable plastic droppers
► Each pouch contains one cassette with one desiccant bag
► Instruction for use
Materials Required But Not Provided
► Timer or stopwatch
► Biohazard disposal container
► Blood collection devices, for the testing of venous whole blood, serum or plasma
► Disposable gloves
For finger stick samples, the following materials are required:
Sterile gauze or cotton
Sample Collection and Test Preparation
Finger stick Specimens(Whole Blood)
► Clean the area to be lanced with an alcohol pad.
► Squeeze the end of the fingertip and pierce it with a sterile lancet.
► Wipe away the first drop of blood with sterile gauze or cotton; collect the sample from the second drop.
► Use dropper to obtain appropriate amount of fresh blood and dispense into the sample well.
Finger stick whole blood should be used immediately after collection.
Serum/Plasma specimens:fresh serum or plasma specimen can be used. No special patient preparation required.
1.Collect whole blood into a collection tube(containing EDTA, Na-citrate or heparin) by venipuncture.
2.Separate the plasma by centrifugation.
1.Collect whole blood into a collection tube(containing no anticoagulants) by venipuncture.
2.Allow the blood to clot.
3.Separate the serum by centrifugation.
Any visible particulate matter in the specimen should be removed by centrifugation or filtration. Avoid using of hemolytic, turbid, microorganism contaminated specimens or specimens stored for over 5 days at 4°C.Specimen should be stored frozen at-20°C for maximum 3 months. Avoid specimen deterioration by multiple freeze-thaw cycles.
Venous Whole Bloods
Venous whole blood can be used immediately after collection or stored up to 4 days at 2-8°C.
Interpretation of Results
► Negative: No apparent band in the test region (1 and 2), only one red band appears in the control region (C). This indicates that no HIV1/2 antibodies have been detected.
► Positive: In addition to the band in the control region (C), other one or two red bands will appear in the test region (1 and 2). This indicates that the specimen contains HIV1/2 antibodies.
► Invalid: If no band appears in the control region(C), regardless of the presence or absence of line in the test region (1 and 2). It indicates a possible error in performing the test. The test should be repeated using a new device.
Newscen HIV(1+2) Antibody Rapid Test can be stored at room temperature(4-30℃, do not freeze) for 24 months from the date of manufacture.
Keep the test cassette in sealed pouch until use. Once you have taken the test cassette out of the pouch, perform the test as early as possible(within 1hour) to avoid test cassette from becoming moist. Do not use the test beyond the indicated expiration date.
The diluent buffer should be stored at room temperature(4-30℃, do not freeze) .
For Invitro Diagnostic Use ONLY
Read the package insert completely before use. It is very important that the correct procedure is followed. Fail to add the patient sample may lead to a false negative result (i.e.a missed positive) .
Contact Person: Mr. Randy.Zhang