Professional Manufacturer of Rapid Test Devices
|Place of Origin:||China|
|Certification:||ISO9001, CE, TUV, FDA|
|Minimum Order Quantity:||10 Boxes or 400 Kits (40 Kits/Box)|
|Price:||Negotiable as per Order Quantity|
|Packaging Details:||40 Kits/Box, 50 Boxes/Carton, 317g/Box, 18kgs/Carton, Box Size: 250x125x65mm, Carton Size: 650x510x330mm|
|Delivery Time:||8 days|
|Payment Terms:||T/T, Western Union, MoneyGram|
|Supply Ability:||100,000 Kits Per Day|
|Format:||Cassette / Strip||OEM/ODM:||Available|
|Usage:||For In Vitro Diagnostic Use Only||CE Certificate:||Yes|
Morphine Drug Rapid Test Kit,
MOR Drug Rapid Test Kit
The MOR Gold Rapid Screen Test Cassette is a rapid qualitative competitive binding immunoassay for determination of Morphine and their metabolites in urine.
Summary and Explanation of The Test
The MOR Gold Rapid Screen Test Cassette is a fast and easy to read membrane chromatographic immunoassay with no need for instrument. The method employs a unique combination of monoclonal and polyclonal antibodies to specifically identify Morphine in urine with a high degree of sensitivity.
Morphine, one of opiates’ derivatives, is central nervous system stimulants that produce alertness, wakefulness, increased energy, reduced hunger, and an overall feeling of well-being. Morphine has been the preferred drug for the management of pain in cases of advanced cancer and other severe conditions.
Morphine and other Opiates’ metabolites may be detected in urine as a result of heroin morphine, codeine, or poppy seed intake. Administration of large doses of Morphine can cause tolerance and physiological dependence in users, and can lead to substance abuse. D-Morphine is controlled substances. GC/MS or immunoassay method has been developed for the determination of 300ng/ml for Morphine.
Reagents And Materials Provided
1. Test Device
A pouched cassette contains a single test for Amphetamine
A transfer pipette seal in foil pouch together with test device
3. Operating Instruction
Materials Required But Not Provided
1. Clock or Timer
2. A container for specimen collection
Warning and Precautions
1. For in vitro diagnostic use only.
2. Do not use kit beyond the expiration date.
3. Urine specimens may be infectious; properly handle and dispose of all used reaction devices in biohazard container.
Sample Collection and Preparation
Collect a fresh urine sample in a clean, dry container, either plastic or glass, without any preservatives. Urine specimens may be refrigerated (2-8℃) and stored up to 72 hours before testing. Urine samples exhibiting visible precipitates should be filtered, centrifuged or allowed to settle. Use only clear aliquots for testing.
1. Bring the urine sample and test components to room temperature if refrigerated.
2. When ready to test, open the pouch at the notch and remove the test device. Place the test device on a clean, flat surface.
3. Fill the urine dropper with specimen. Holding the dropper vertically, dispense 2-3 drops (about 70-100µl) of urine without air bubble into the sample well.
4. Read the result in 5-10 minutes.
Interpretation of Results
1. Negative: Two pink color bands appear. This indicates MOR in the specimen is below the detection sensitivity (300ng/ml).
2. Positive: Only one pink band for control appears. This indicates the MOR in the specimen is above the detection sensitivity.
3. Invalid: The presence of a control line is necessary to validate test performance. If no band appears the test should be repeated with a new test device.
Note: There is no meaning attributed to color intensity for C and T lines.
Contact Person: Mr. Randy.Zhang