Professional Manufacturer of Rapid Test Devices
|Place of Origin:||China|
|Certification:||ISO9001, CE, TUV, FDA|
|Model Number:||COVID-19 IgG/IgM|
|Minimum Order Quantity:||10 Boxes or 400 Kits (40 Kits/Box)|
|Price:||Negotiable as per Order Quantity|
|Packaging Details:||40 Kits/Box, 50 Boxes/Carton, 317g/Box, 18kgs/Carton, Box Size: 250x125x65mm, Carton Size: 650x510x330mm|
|Delivery Time:||8 days|
|Payment Terms:||T/T, Western Union, MoneyGram|
|Supply Ability:||50,000 Kits Per Day|
|Methodology:||Colloidal Gold||OEM / ODM:||Available|
Colloidal Gold Coronavirus Test Kit,
20min Coronavirus Test Kit,
Serum Plasma Coronavirus Test Cassette
Novel Coronavirus (2019-nCoV) Antibody IgG/IgM Rapid Test Cassette
COVID-19 Rapid Teste Kit Novel Coronavirus Colloidal Gold Antibody IgG/IgM Test Cassette
|1||Fast to get the result in 15-20 mins|
|2||High sensitivity and specificity|
|3||Simple to use|
|4||High Accurate (97.51%) and Reliable (Credibility interval: 95.79%～98.67%)|
|5||Certified by Authoritative Certification|
|7||Unique 3-line Patented Design|
Meet Multiple Criteria as follows:
EN ISO 13485:2016 EN 13612:2002
EN ISO 14971:2012 EN ISO 23640:2015
EN ISO 18113-1:2011 EN ISO 18113-2:2011
EN ISO 15223-1:2016 EN13641:2002
COVID-19IgG/lgM Fast Diagnostic Cassette is for the detection of lgG/lgM in human serum, plasma or whole blood.
It is used as a supplementary test indicator for suspected cases of Novel Coronavirus negative nucleic acid detection or used combine with nucleic acid detection in the diagnosis of suspected cased, It cannot be used as the evidence for the diagnosis and exclusion of COVID-19.
Materials Required But Not Provided
► Timer or stopwatch
► Blood collection devices, for the testing of venipuncture whole blood, serum or plasma
► Biohazard disposal container
► Disposable gloves
► Pipette and pipette tip
For fingerstick samples, the following materials are required:
Sterile gauze or cotton
Cassette and Materials Provided
► Test cassette(individually pouched)
► Each pouch contains one cassette with one desiccant bag
► Diluent buffer
► Instruction for use
► Plastic dropper(optional)
Interpretation of Results
► Positive of lgG and IgM: if three lines appear in their respective zone C, G, M, then it Indicates IgG and IgM test result is positive.
► Positive of lgM: if two lines respectively appears in the line of Zone C and M, It indicates IgM test result is positive.
► Positive of IgG: if two lines respectively appears in the line of Zone C and G, It indicates lgG test result is positive.
► Negative of lgG and IgM: if Only one color line in the control zone C it indicates IgG and IgM test result is negative.
► Invalid: If no color line appears in the control zone C, then the test is invalid. Discard the test cassette and resume the performance with new cassette through correct procedures.
Sensitivity and Specificity
A total of 523 clinical samples were tested for validation, of which 142 were positive and 381 were negative. The test results are as follows:
|NewScen||Clinical Sample Information||Total|
Positive Coincidence Rate
Positive Coincidence Rate: A/(A+C)%=90.85% (Credibility Interval: 84.85% ～95.04%)
Negative Coincidence Rate: D/(B+D)%=100.00% (Credibility Interval: 99.04%～100.00%)
Total Coincidence Rate: (A+D)/( A+B+ C+D)% = 97.51% (Credibility interval: 95.79%～98.67%)
IgG/lgM Rapid Test Cassette has a built-in procedural control to see whether or not the operation is correct or effective, if one color line appeared on the control zone C, it indicates that the test operation is Correct.
COVID-19 IgG/IgM Rapid Diagnostic Kit should be placed in dark place at 2-30°C for 12 months from the date of manufacture.
Keep the test kit in sealed pouch until use. Once you have taken the test cassette out of the pouch, use it immediately. Do not use the test kit once it exceeds the indicated expiration date.
Contact Person: Mr. Randy.Zhang